[Cytometry] BC Navios or BD FACSCanto II ?
jakso.pal at pte.hu
Sat Feb 1 02:56:19 EST 2014
Thanks for the informations. I'll check the publications.
I have questions:
How the UV light is preserved? Do you mean that the Navios is better detetecting parameters from violet laser?
What do you mean on the excellent photoprotection? The Canto is also equiped with a dark housing for the carousel.
Am I wrong that in the Navios only the lasers and the excitation optics are "air conditioned" but the flow cell not ?
2014.01.31. dátummal, 16:17 időpontban marie c bene <mariecbene at gmail.com> írta:
> Hi Pal
> Both Canto II and Navios are good instruments and you might be interested
> in a paper our French group published recently about harmonization between
> them (Solly F et al. Cytometry part A)
> Navios differs by a better discrimination of fluorescence at low
> intensities linked to the higher number of bits (easily demonstrated by the
> broad CVs of dim beads or cells in the first two decades on a Canto), by
> providing 12 parameters (10 FL), being equipped with a carousel with single
> tube vortexing before acquisition (important for large batches or long
> acquisition times i.e. single platform absolute counts or minimal residual
> disease , see Lambert C Cyometry part B, 2005), excellent photoprotection
> for fragile fluorochromes, very stable thermostated flow cell and lasers),
> and finally no light loss with UV.
> Hope this will help you.
> MC Béné
> Cytometry mailing list
> Cytometry at lists.purdue.edu
> Search the list archive at http://tinyurl.com/cytometry
Magyarország első egyeteme – 1367
More information about the Cytometry