[Cytometry] Question about FACSAria daily shutdown solution
marie.follo at uniklinik-freiburg.de
Mon Jan 16 03:51:26 EST 2012
Thanks for bringing this topic up.
My institute has started recommending that people switch over from
using 70% Ethanol to 60% Isopropanol for general disinfection
purposes. I don't know of any inherent disadvantages to using the
Isopropanol on machine components, but would love to hear from anyone
if there are any.
The advantage of this would be that it gets you away from the problem
of having to choose between having the additives present in the
Ethanol vs. the high cost of Ethanol without the additives. I haven't
tried the Isopropanol yet, but was seriously considering the switch
for exactly those reasons.
Am 12 Jan 2012 um 20:58 hat Schmitt, Steffen geschrieben:
> Dear colleagues,
> Happy new year to everybody and interesting discussions on this list!
> What I am interested in is, which shut down solutions you use for
> routinous daily shut down of the cell sorters FACSAria.
> We have 3 of them - all at least at Aria IIu level.
> We use 80% Ethanol (which originally contains 1% Petrolether) in
> deionised water.
> After now more than 1,5 years of running these machines (2-3 sorts per
> day on each instrument), we faced significantly reduced FSC/ SSC
> signals at all 3 instruments at once. On one instrument we could see
> mouse bone marrow only with logarithmic amplification of FSC/ SSC -
> doublet discrimination wasn´t possible at all.
> Cleaning the cuvette with FACSClean, FACSRinse and Ethanol didn´t help
> much. At the end with Neodisher (a more acidic cleaning solution) we
> were able to get back to acceptable scatter signals.
> After discussions with the company the focus lies now on the
> petrolether in our shut down solution.
> Now I want to hear your experiences, problems and solutions.
> Thank you all for your replies.
> Dr. Steffen Schmitt
> Head of the Flow Cytometry Unit
> German Cancer Research Centre
> (Deutsches Krebsforschungszentrum in der Helmholtz-Gemeinschaft,
> Stiftung des öffentlichen Rechts)
> Microscopy Core Facility; Flow Cytometry Unit;
> KST: W220
> Im Neuenheimer Feld 280
> D-69120 Heidelberg
> Phone: +49-(0)6221-42-1261 (office)
> +49-(0)6221-42-3643 (lab)
> Fax: +49-(0)6221-42-3755
> E-Mail: steffen.schmitt at dkfz-heidelberg.de
> Web: www.dkfz.de/en/facs/index.html
> Confidentiality Note: This message is intended only for the use of the
> named recipient(s) and may obtain confidential and/ or privileged
> information. If you are not the intented recipient, please contact the
> sender and delete the message. Any unauthorized use of the information
> contained in this message is prohibited.
> Cytometry mailing list
> Cytometry at lists.purdue.edu
> Search the list archive at http://tinyurl.com/cytometry
Marie Follo, Ph.D.
Abt. Innere Medizin I
Zentrale Klinische Forschung
Tel.: +49 761 270 63770
e-mail: marie.follo at uniklinik-freiburg.de
More information about the Cytometry