[Cytometry] APC problems
derek.davies at cancer.org.uk
Fri Feb 27 14:59:09 EST 2009
What flavo(u)r of red laser do you have? If its a 648nm you may be picking
up direct laser light which would sort of explain why you get better
separation by moving the APC filter up a tad. Don't forget that bandpass
filters will let light in from outside their range - it may not be much but
if its high power laser it will reduce your sensitivity. We have had to be
very careful with optical filter selection on both my LSRs.
On 26/2/09 12:27, "Christian Willberg" <Chris.Willberg at ndm.ox.ac.uk> wrote:
> Dear All,
> I was hoping somebody could help us with a problem we're having with
> our APC PMT on a new LSRII, our local BD engineer is at a loss.
> We have been struggling to get good APC stains with antibodieswe know
> work very well on other machines. When we checked with BD rainbow
> beads we noticed that the APC dim beads where bunching up halfway
> along the axis - this wasn't seen on either the Alexa700 PMT or APC-
> Cy7 PMT (please see attached PDF). We checked that this wasn't due to
> the filter (660/20) using a second 660/20 filter and the same pattern
> was seen. However, when we swapped the 660/20 filter for a 670/14 -
> the problem was resolved. Ideally we would like to use the 660/20
> filters for APC and not the 670/14. Can anyone explain what could be
> causing this and how we can fix it?
> Many thanks,
> Chris Willberg
> Chris Willberg
> Biomedical Research Centre - Immunology Theme
> University of Oxford
> Oxford, UK
> Tele: 01865 271290
> or tele: 01865 222352
> e-mail: Chris.Willberg at ndm.ox.ac.uk
> Cytometry mailing list
> Cytometry at lists.purdue.edu
This communication is from Cancer Research UK. Our website is at www.cancerresearchuk.org. We are a charity registered under number 1089464 and a company limited by guarantee registered in England & Wales under number 4325234. Our registered address is 61 Lincoln's Inn Fields, London WC2A 3PX. Our central telephone number is 020 7242 0200.
This communication and any attachments contain information which is confidential and may also be privileged. It is for the exclusive use of the intended recipient(s). If you are not the intended recipient(s) please note that any form of disclosure, distribution, copying or use of this communication or the information in it or in any attachments is strictly prohibited and may be unlawful. If you have received this communication in error, please notify the sender and delete the email and destroy any copies of it.
E-mail communications cannot be guaranteed to be secure or error free, as information could be intercepted, corrupted, amended, lost, destroyed, arrive late or incomplete, or contain viruses. We do not accept liability for any such matters or their consequences. Anyone who communicates with us by e-mail is taken to accept the risks in doing so.
More information about the Cytometry