[Cytometry] 5 color compensation issue
Haviland, David L
David.L.Haviland at uth.tmc.edu
Tue Feb 3 10:23:12 EST 2009
Yes. If you compare the emission profile of PE-Cy5.5 to PE you are going to find overlapping emission spectra that can't be compensated. A better choice for your user would have been PE-Cy5. I had this once myself and kindly asked the user not to bring that combination in my flow room again. It was all made clear to my user when I showed her the emmission profiles of the reagents in question.
Hope this helps...
From: cytometry-bounces at lists.purdue.edu on behalf of Zhugong Liu
Sent: Mon 2/2/2009 9:29 PM
To: cytometry at lists.purdue.edu
Subject: [Cytometry] 5 color compensation issue
We are running 5-color flow on FC500 and have some compensation issues.
The 5-color cocktail includes antibodies with the following
fluorechromes: FITC, PE, PE-Cy5.5, PE-Cy7 and APC. Do you think there
might be some reagent interactions in this panel combo, especially
PE-Cy5.5 and other dyes? Our single controls were well compensated using
the software-generated matrices, however the fully stained samples were
Thanks for any comment/suggestion.
Jason Liu, PhD
Asst. Director, Flow Cytometry
Tolerance Assays and Data Analysis
UCSF/Immune Tolerance Network
3 Bethesda Metro Center, Suite 400
Bethesda, MD 20814
Email: zliu at immunetolerance.org
Cytometry mailing list
Cytometry at lists.purdue.edu
More information about the Cytometry