Validation report for multiple staining
Daniel.Roberts at covance.com
Thu Jun 26 09:34:24 EDT 2008
The big ticket item here in the states is the FDA's 21st Code of Federal
Regulations (CFR), part 11. This entails computer/system validation to
ensure the integrity of electronic data. I'm not sure if the Canadian
Guidelines have something similar for electronic data capture, but that
should be the only hurdle. As far as a Method Validation goes...if you run
your protocol - no matter how many colors - and perform a comparison to
microscopy or other gold standard technique for your endpoint, as long as
the governing scientific body of your lab signs off on it, you should be
good to go. I had IT QC, Management, and Authors/testers sign off on my
documents, with QA giving the final approval.
I've validated 2 flows, a BD Calibur, and a BD Canto, and most of the
validation work is spend on the electronic data / system security portions.
Your QA department should have some sort of format for dealing with GLP
automation - unless the flow is their first piece of equipment that will
generate data. Just so you know, unless you are going to submit the data for
an IND or NDA, I do not believe that you need to go the FDA GLP 21 CFR part
11 route. Here's a website though if it will help:
Also, if you share details of the method you're validating, I'm sure that
some of the experts on this distribution list will have some pointers on
which antibodies / gating logic is more appropriate.
Technical Resource Specialist
Genetic Toxicology Department
Covance Laboratories, Vienna, VA
703-245-2200 x5136 (Office)
From: Cythéris - Stéphanie BEQ [mailto:sbeq at cytheris.com]
Sent: Tuesday, June 24, 2008 12:36 PM
Subject: Validation report for multiple staining
I have just join a private company (I was previously working in the
academic) and one of my topics is to install flow cytometry (not a problem
for research use) by Quality Assurance (that is the problem) and I'm not
familiar with QA. I have already found the Canadian Guideline for flow
cytometry (2001) and the H42-A2 from NCCLS on the web and the daily/monthly
QC of the instrument itself should not represent a big difficulty, but the
validation of staining, antibodies panel, compensation, multicolor analysis,
gating, reproducibility, what kind of calibrator should I choose.....remains
unclear for me.
Is anybody having a two (or more of course!) colors staining validation
report (or validation plan)? It could be really helpful for me to adapt it
and develop it for an 8 colors panel.
Does more recent guidelines exist (FDA or CDC or else?) ?
Thank you for your help.
Immunological Studies Manager
cid:image001.gif at 01C8B782.D5D90420
175 rue Jean-Jacques Rousseau
92130 ISSY-LES-MOULINEAUX, France
sbeq at cytheris.com <mailto:bassouline at cytheris.com>
Tél: 33(0)1 5888 3819
fax 33(0)1 4644 4446
STATEMENT OF CONFIDENTIALITY
The information contained in this electronic message and any attachments to
this message are intended for the exclusive use of the addressees and may
contain confidential or privileged information. If you are not the intended
recipient, please notify "Cytheris" immediately at either (33)15888 3800 or
at info at cytheris.com <mailto:info at cytheris.com> , and destroy all copies of
this message and any attachments.
Confidentiality Notice: This e-mail transmission
may contain confidential or legally privileged
information that is intended only for the individual
or entity named in the e-mail address. If you are not
the intended recipient, you are hereby notified that
any disclosure, copying, distribution, or reliance
upon the contents of this e-mail is strictly prohibited.
If you have received this e-mail transmission in error,
please reply to the sender, so that we can arrange
for proper delivery, and then please delete the message
from your inbox. Thank you.
-------------- next part --------------
HTML attachment scrubbed and removed
-------------- next part --------------
A non-text attachment was scrubbed...
Size: 3265 bytes
Desc: not available
Url : https://lists.purdue.edu/pipermail/cytometry/attachments/20080626/70f90898/image001-0005.gif
More information about the Cytometry