Macrosort Head for B.D. FACS VANTAGE
trotter at scripps.edu
Thu Aug 13 15:09:10 EST 1998
The little set screw will often "push" the cannula off
center, so I would suggest *carefully* re-centering it with enough
tension to hold it without moving it off center. There is a little
"Kentucky windage" involved, but you should be able to get it.
I doubt the cannula is bent, and am guessing the set screw is
a tad too tight.
On Thu, 13 Aug 1998, Keyvan Keyvanfar wrote:
> The sample insertion rod in my Macrosort head on one of my FACS
> VANTAGES is off center. I do know that the insertion rod is not bent. I
> was wondering if it is possible to align the insertion rod within the
> Macrosort head?
> Any suggestions, beside buying a new one!
Director, Flow Cytometry
The Scripps Research Institute
10666 North Torrey Pines Rd.
La Jolla, California 92037
More information about the Cytometry