A clean machine -Reply
fharrod at pharmingen.com
Tue Feb 11 20:22:40 EST 1997
Reply to: RE>>A clean machine -Reply
We've been using a dilute sodium hydroxide solution to clean the tubing in our three
FACscans for several years now. We used to use bleach, as recommended by the
manufacturer, and sometimes we would have clogs which never seemed to clear up
completely. Our protein chemist recommended NaOH to me late one afternoon when
I was particularly frustrated with a stubborn clog. The NaOH cleared it right
out, and from then on we've used it regularly to keep the lines clear with few
problems. I've always wondered if anyone else has tried it and how it has
worked for them.
Date: 2/11/97 4:21 PM
From: Jim Zanghi
>Small beads staying in the instrument can be a problem. Triton X-100 at
>0.1% (the same concentration used to permeabilize cells) in water will
>remove most of them.
>> Running ethanol
>> after the beads seems to help push them through more quickly.
>> Tony Bakke
I was wondering about what types of solutions can be run through a flow
cytometer without causing damage or corrosion. Is pure ethanol okay? On
the advise of others, I have run 50% hot bleach to clear up clogs, but I
was very weary about doing this since bleach corrodes stainless steel.
I've always followed with a 5 minute water rinse, but this concentration of
bleach seems a bit extreme (we routinely use 10% bleach at room temp).
If 0.1% triton or 70% ethanol is as effective as 50% hot bleach, than this
seems to be the way to go. Any thoughts about this? What do
James A. Zanghi
Dept. of Chemical Engineering
------------------ RFC822 Header Follows ------------------
>From flowcyt.cyto.purdue.edu!owner-cyto-sendout Tue Feb 11 14:08:37 1997 remote
Received: from flowcyt.cyto.purdue.edu by donews.cts.com with smtp
(Smail184.108.40.206 #5) id m0vuQNO-00005zC; Tue, 11 Feb 97 14:08 PST
Received: by flowcyt.cyto.purdue.edu (940816.SGI.8.6.9/930416.SGI.AUTO)
for cyto-sendout id MAA10161; Tue, 11 Feb 1997 12:30:01 -0500
Received: from merle.acns.nwu.edu by flowcyt.cyto.purdue.edu via ESMTP
for <cytometry at flowcyt.cyto.purdue.edu> id LAA09604; Tue, 11 Feb 1997 11:18:55
Received: from [220.127.116.11] (barney.chem-eng.nwu.edu) by merle.acns.nwu.edu
(18.104.22.168/16.2) id AA138027530; Tue, 11 Feb 1997 10:12:11 -0600
Content-Type: text/plain; charset="us-ascii"
Date: Tue, 11 Feb 1997 10:19:08 -0600
From: zanghi at merle.acns.nwu.edu (Jim Zanghi)
Subject: Re: A clean machine -Reply
More information about the Cytometry